A laboratory bench-top system for automatically determining annealing point temperature and strain point temperature of a glass fiber below 800C.
Operated according to ASTM C-336 Method A or modified version of this procedure.
- Bench top instrument console
- 800°C furnace
- programmable process controller
- measuring head with LVDT sensor
- type "S" thermocouple
- 1Kg load weight
- computer software
- accessories and instruction manual. (Computer not included.)
|Temperature Range||RT to 800°C|
|Heated Length||10.75 inches (273 mm)|
|LVDT Displacement Range||±0.500 inch (±12.7 mm)|
|LVDT Sensitivity||0.7 mV/V/.001 inch(28 mV/V/mm)|
|Sample Size||508 mm long x 0.65 mm diameter|
|Power Requirements||120vac/10 amp or 240vac/5amp, 50 or 60hz|
|Computer Requirements||Desktop with available PCI slot|
|Temperature Control||Honeywell UDC 3200 Process controller|
|Computer Interface Software||Included|
|SRM Calibration samples||Optional|
|Bench-top Footprint (stand)Length x Depth x Height||12 x 12 x 33 (31 x 31 x 84 mm)|
|Bench-top Footprint (controller)Length x Depth x Height||18 x 12 x 5 (46 x 31 x 13 cm)|
FROM THE BLOG...
Orton has partnered with Zonos and United Parcel Service so Customers around the world can order small...
06/23/2020Shelf Life of Cones
Do cones have a shelf life? At the Edward Orton Jr. Ceramic Foundation, a common question asked to...
05/12/2020New Test Equipment Installed
Orton's Materials Testing Service Department recently added a new laser flash machine to measure thermal...