A laboratory, bench-top, fiber elongation system for automatically determining Littleton softening point of a glass fiber below 900c
Operated according to ASTM C-338 Method or modified version of this procedure.
- Bench top instrument console
- 900°C furnace
- programmable process controller
- laser micrometer and controller
- type "S" thermocouple
- computer software
- accessories and instruction manual. (Computer not included.)
|Temperature Range||RT to 900°C|
|Furnace Core||Stainless Steel|
|Heated Length||3.25 inches (83 mm)|
|Laser Type||FDA Class II / IEC Class 1|
|Laser Micrometer Range||0 - 35 mm|
|Sample Size||235 mm long x 0.65 mm diameter|
|Power Requirements||120vac/10amp or 240vac/5amp, 50 or 60 Hz|
|Temperature Control||Honeywell UDC 3200 Process controller|
|Computer Interface Software||Included|
|SRM Calibration samples||Optional|
|Bench-top Footprint (stand) Length x Depth x Height||10 x 10 x 15 (25 x 25 x 38 mm)|
|Bench-top Footprint (controller) Length x Depth x Height||18 x 12 x 5 (46 x 31 x 13 cm)|
FROM THE BLOG...
Orton has partnered with Zonos and United Parcel Service so Customers around the world can order small...
06/23/2020Shelf Life of Cones
Do cones have a shelf life? At the Edward Orton Jr. Ceramic Foundation, a common question asked to...
05/12/2020New Test Equipment Installed
Orton's Materials Testing Service Department recently added a new laser flash machine to measure thermal...